Payment & Shipping Terms:
|Workspace Volumeial::||W 1000 × H 1000 × D 1000 Mm||Controller::||LCD Touch Panel|
|Ramping And Soaking Rate:||-20℃↑ +100℃ Within35 Mins (no-load) +20℃↓ -20℃ Within 45mins (no-load)||Exterior Size::||W 1200 × H 2035 × D 2015 Mm|
|Temperature Range:||Single Stage Refrigeration System||Humidity Range:||20%R.H～98%R.H|
1000 L Single Door Refrigerant 404A Air Cooled Humidity And Temperature Controlled Chamber
Able to accurately simulate a wide range of complicated natural environments, and is suitable for
reliability test in industrial products. Meet GB5184.108.40.206.6-95 standard requirements of environmental
testing equipment and test methods for the basic parameters of electric and electronic products under
the condition of humidity, low temperature, high temperature, and constant heat.
Applicable to environmental adaptability and reliability test in such industrial units as electronics, electrical
appliance, battery, plastics, food, paper product, vehicle, metal, chemistry, building material, research
institution, inspection and quarantine bureau, university etc..
|Workspace volume||W 1000 × H 1000 × D 1000 mm|
|Exterior size||W 1200 × H 2035 × D 2015 mm|
|Temperature range||-20℃ ～ +150℃|
|Ramping and soaking rate||
-20℃↑ +100℃ within 35 mins (no-load)
+20℃↓ -20℃ within 45 mins (no-load)
|Temperature and humidity performance measurement comply with related regulation of IEC60068-3-5 standard; Sensors placed in the air outlet.|
Overall structure and chamber was composed of three parts as below.
Insulation box, separate refrigeration units, and electrical control cabinet.
wall material: high-quality carbon steel with static color spray
inner wall material: SUS304 # matte stainless steel plate
Insulation materials: rigid polyurethane foam insulation layer + glass fiber.
|Door||Heating wire was installed at the door frames to prevent condensation at low temperatures.|
|Observation window||With 1 set 460 × 560mm (width × height) observation window, multi-hollow electric insulation coated glass prevent condensation effectively|
|Cable port||Φ50mm*2 located on both sides (each*1) with rubber stopper and plastic cover|
|Lighting device||1 *located on observation window|
Quality nickel-chromium alloy wire electric heaters,
Non-contact control mode (SRR).
Steam heating and humidification method;
Stainless steel sheathed heater;
Heater control: contactless control (SSR);
Water level control device, heater anti-dry device.
|Water outlet hole||Available for drain the condensate water|
|Sample holder||Two layers of stainless steel sample holder.|
|Mobile Casters||Mobile Casters *4 with foot cups|
|Electric control box||Total power circuit breaker, over-temperature protection.|
|Water supply system||Water pump automatic supply|
Environmental test chamber Standard
GJB150.3 high-temperature testing
GJB150.4 low-temperature testing
GJB150.9 humid testing
GB11158 Technical conditions for high-temperature testing box
GB10589-89 Technical conditions for low-temperature testing box
GB10592-89 Technical conditions for high and low temperature testing box
B/T10586-89 technical conditions for humidity testing chamber
GB/T2423.1-2001 Low-temperature testing methods
GB/T2423.2-2001 High-temperature testing methods
B/T2423.3-93 Testing Method for humidity Test Chamber
GB/T2423.4-93 Alternating hot and humid testing method
GB/T2423.22-2001temperature testing method
IEC60068-2-1.1990 low-temperature testing method
IEC60068-2-2.1974 high-temperature testing method
We will send you operating instructions, English video by Email. Or you can download on our
officail website directly.
Most of our machine is shipped with a whole part, which means it is already installed, all you
need to do is to connect the power cable and start to use. And if possible, we can also install
your machine at site.